Company: Hologic, Inc. Date of Enforcement Report: 12/27/2018 Class II PRODUCT I-View Contrast Enhanced Digital Mammography, ASY-08109 Recall Number: Z-0685-2019 REASON Calibration issue not possible to visualize contrast uptake in the subtracted mammography images RECALLING FIRM/MANUFACTURER Hologic, Inc., Danbury, CT on 12/6/2018. Voluntary: Firm Initiated recall is ongoing. VOLUME OF PRODUCT IN COMMERCE 337 units...Read More
Company: Elekta, Inc. Date of Enforcement Report: 12/27/2018 Class II PRODUCT MOSAIQ Oncology Information System Version 2.64. MOSAIQ is an oncology information system used to manage workflows for treatment planning and delivery Recall Number: Z-0686-2019 REASON If a Setup field (CT, kV, MV or MVCT) is created by COPYING a treatment field or another setup...Read More
Company: Radiometer America Inc Date of Enforcement Report: 12/22/2018 Class II PRODUCT ABL800 analyzer, model numbers 393-800 and 393-801. Recall Number: Z-0682-2019 REASON The analyzer software may cause a mis-match of patient demographics and test results when a certain sequence of events occur. RECALLING FIRM/MANUFACTURER Radiometer America Inc, Brea, CA on 7/3/2018. Voluntary: Firm Initiated...Read More
Company: Medtronic Inc. Date of Enforcement Report: 12/22/2018 Class II PRODUCT Medtronic MiniMed 630G 6025805-003_C, Rx Only Medtronic MiniMed 670G 6025805-005_B, Rx Only Medtronic MiniMed 640G (International) Recall Number: Z-0683-2019 REASON Reports of occurrences in which insulin pumps with version 4.10 software have failed to make expected audio sounds during alerts, alarms or sirens. This...Read More
Schedule Discussion with John F. Murray, Jr.
John is currently providing telephone and face-to-face meetings to discuss: Cybersecurity, Part 11, 483 Response, design controls expectations for software documentation, and other topics.
Leave a message and we will contact you to schedule: