Company: Ortho-Clinical Diagnostics Date of Enforcement Report: 12/20/18 Class II PRODUCT VITROS 5,1 FS Chemistry System, Version 3.0 & below For use in the in vitro quantitative measurement of a variety of analytes of clinical interest in certain biological fluids (depending on the analyte) such as serum, plasma, urine and cerebrospinal fluid. Recall Number: Z-0651-2019...Read More
Company: Ortho-Clinical Diagnostics Date of Enforcement Report: 12/20/18 Class II PRODUCT VITROS 5600 Integrated System, Version 3.3.1 & below For use in the in vitro quantitative, semi-quantitative, and qualitative measurement of a variety of analytes of clinical interest, using VITROS Chemistry Product Slides, VITROS Chemistry Products MicroTip Reagents and VITROS Immunodiagnostic Products Reagents. Recall Number:...Read More
Company: Philips Electronics North America Corporation Date of Enforcement Report: 12/14/2018 Class II PRODUCT Xper Flex Cardio Physiomonitoring System, Software versions 1.5.16.0322 and 1.6.1.0390, Model Numbers 453564241901, 453564241911, 453564483321, 453564483331, 453564621791, 453564621801, 453564669081 Recall Number: Z-0614-2019 REASON There are 3 issues: 1) Under certain conditions, real time waveforms may not be accurately plotted or displayed....Read More
For devices where the medical device manufacturer intends to maintain remote access capability for service and updates, controls are needed to ensure this access does not compromise the security or operation of the device. There is one FDA recognized standard that addresses this for clinical laboratory devices, but many of the provisions could be instructive...Read More
Schedule Discussion with John F. Murray, Jr.
John is currently providing telephone and face-to-face meetings to discuss: Cybersecurity, Part 11, 483 Response, design controls expectations for software documentation, and other topics.
Leave a message and we will contact you to schedule: