Company: Hologic, Inc. Date of Enforcement Report: 12/27/2018 Class II PRODUCT I-View Contrast Enhanced Digital Mammography, ASY-08109 Recall Number: Z-0685-2019 REASON Calibration issue not possible to visualize contrast uptake in the subtracted mammography images RECALLING FIRM/MANUFACTURER Hologic, Inc., Danbury, CT on 12/6/2018. Voluntary: Firm Initiated recall is ongoing. VOLUME OF PRODUCT IN COMMERCE 337 units...Read More
CSV Training Course
Learn FDA expectations for software validation for computer systems, quality system software, manufacturing and production process software, and engineering tools. Email email@example.com for more info.