Company: Insulet Corporation
Date of Enforcement Report: 6/10/2020
Omnipod DASH Insulin Management System (mg/dL configuration), Catalog Number PT000010 M/D: INT1D001MG – Product Usage: is intended for subcutaneous delivery of insulin at set and variable rates for the management of diabetes mellitus in persons requiring insulin.
Recall Number: Z-2164-2020
After the device has been in use for about 2 months, data processing in the PDM can be slowed such that the Bolus Calculator fails to accurately subtract the correct amount of IOB before suggesting a bolus amount.
Insulet Corporation on 4/2/2020. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE