Company: Abbott Laboratories Inc. (St Jude Medical) Date of Enforcement Report: 6/5/2019 Class II PRODUCT TactiSys Quartz Equipment (with software version 1.7) Model: PN-004 400 Part Numbers: 600043767 (China), 100154367 (Global), 100154368 (SWAP) Recall Number: Z-1493-2019 REASON In reported cases, the device log on the TactiSys Quartz Equipment operating on Software Version 1.7.0 fills the...Read More
Company: Change Healthcare Israel Ltd. Date of Enforcement Report: 6/5/2019 Class II PRODUCT Horizon Cardiology Hemo: Software version 11.x and 12.x Recall Number: Z-1489-2019 Software version 13.x Recall Number: Z-1490-2019 REASON users are not notified of procedure medication discrepancies between the Vitals and Meds, the Procedure Notes and Patient Common Data screens in Horizon/McKesson Cardiology...Read More
CSV Training Course
Learn FDA expectations for software validation for computer systems, quality system software, manufacturing and production process software, and engineering tools. Email firstname.lastname@example.org for more info.