Company:Tandem Diabetes Care Inc.
Date of Enforcement Report 7/25/2018
t:slim G4 Insulin Pump with Dexcom G4 Platinum CGM Software version: 22.214.171.124, Firmware version: 004722, Firmware, Fuel Gauge, Binary Rev B.
Recall Number: Z-2471-2018
The fuel gauge, the component that reads and reports the battery parameters to the pump, could provide inaccurate readings, which present to the user in one of two ways: A) by triggering a Malfunction 4 Alarm, or B) by triggering a succession of notifications prior to the pump shutting off, including Alert 2 (Battery Low), Alert 3 (Battery Very Low) and Alarm 12 (Battery Very Low). The alarms in both scenarios notify the user that the pump has stopped delivering insulin.
Tandem Diabetes Care Inc., San Diego, CA on 4/23/2018. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
U.S. Consignees: IN, AZ, UT, TX, NY, NJ, MD, MN, IL, CO, AL, WI, CA, PA, OH, OR, MS, FL, NM, ID, NV, KY, VA,