Company: Steris Corporation
Date of Enforcement Report: 7/18/2018
Reliance 1227 Cart and Utensil Washer/Disinfector, FW03101, FW03102, FW03S003 Product Usage: The Reliance 1227 Cart and Utensil Washer/Disinfector is intended for use in the cleaning and low-level disinfection of bedpans and urinals, basins, carts, beds, theatre shoes and other miscellaneous reusable items used in the care of patients.
Recall Number: Z-2392-2018
The firm has become aware that the Reliance 1227 Cart and Utensil Washer/Disinfector s Chemical Low Level alarm, intended to prevent the user from initiating a cycle when a low chemical level situation occurs, does not operate as intended. Currently, if a low chemical level situation occurs, the alarm will only be generated at the unit s next power-up or when accessing service mode. The failure of the Reliance 1227 to identify that the chemicals used for cleaning are low or empty could result in bedpans and urinals, basins, case carts, beds, theater shoes and other miscellaneous reusable patient care items not being properly cleaned before disinfection or reuse.
Steris Corporation, Mentor, OH on 5/23/2018. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide and Canada