Company:CME America, LLC.Date of Enforcement Report: 2/25/2015 Class lI: PRODUCT BodyGuard 323 pump, models 100-510PXSI, 100-516PXS, 100-517PXS, 100-518PXS, 100-603XSA and 100-603XSAP Recall Number Z-1126-2015 REASON CME America is recalling the BodyGuard and BodyGuard 323 Infusion pumps due to the potential for an over delivery. RECALLING FIRM/MANUFACTURER CME America, LLC, Golden, CO on 1/16/2015 Voluntary: Firm...Read More
Company: Siemens Medical Solutions USA, Inc Date of Enforcement Report 2/25/2015 Class lI: PRODUCT Artis One; The Artis One is an angiography system developed for diagnostic interventional procedures including, but not limited to pediatric and obese patients. Procedures that can be performed with the Artis One include cardiac angiography, neuro-angiography, general angiography, rotational angiography, multipurpose...Read More
Company:Alivecor SFO.Date of Enforcement Report: 3/4/2015 Class lI: PRODUCT Alive ECG App 2.1.2 ( a medical device application for the Apple iOS operating system, intended to be used with the AliveCor Heart Monitor. The AliveCor Heart Monitor is intended to record, store and transfer single-channel electrocardiogram (ECG) rhythms. The AliveCor Heart Monitor also displays ECG...Read More
CSV Training Course
Learn FDA expectations for software validation for computer systems, quality system software, manufacturing and production process software, and engineering tools. Email email@example.com for more info.