Recipient: American Optisurgical Inc. Product: TX1 Tissue Removal System Date: 6/14/2012 This inspection revealed that these devices are adulterated within the meaning of section 501(h) of the Act [21 U.S.C. § 351(h)], in that the methods used in, or the facilities or controls used for, their manufacture, packing, storage, or installation are not in conformity...Read More
Company:Philips Medical Systems (Cleveland) Inc.Date of Enforcement Report 6/13/12 Class ll: PRODUCT The Brilliance iCT and iCT SP are Computed Tomography X-Ray Systems intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes. These devices may include signal analysis and display equipment, patient, and...Read More
Schedule Discussion with John F. Murray, Jr.
John is currently providing telephone and face-to-face meetings to discuss: Cybersecurity, Part 11, 483 Response, design controls expectations for software documentation, and other topics.
Leave a message and we will contact you to schedule: