Recipient: Visual Telecommunications NetworkProduct: ViTelCare Trutle Home Patient Monitoring Systems Date: 9/13/05 Software used in the Turtle 600 and Turtle 800 devices has not been validated and a risk analysis has not been conducted for the finished devices [21 CFR 820.30(g)]. FDA District: Baltimore District OfficeRead More
Recipient:Guidant CorpProduct: TA-200 Thermal Angel Blood and IV Fluid Infusion Warmer Date:9/22/05 There are no corrective and preventive actions to change design control procedures to identify or prevent memory overflow or similar software programming problems from recurring. Software used as part of production and the quality system has not been fully validated for its intended...Read More
CSV Training Course
Learn FDA expectations for software validation for computer systems, quality system software, manufacturing and production process software, and engineering tools. Email email@example.com for more info.