Company: Baxter Healthcare Corporation
Date of Enforcement Report: 6/30/2022
NaviCare Nurse Call/Voalte Nurse Ca
An issue has been identified with Phillips (Emergin) and Longleaf non relay (Connexall, Vocera, Cerner) wireless integrations used with NaviCare/Voalte Nurse Call, software versions 3.9.100 through 3.9.300. Calls placed from a push button call device, such as a push button switch, call cord, or from the push buttons on a room audio station, will be canceled on the nurse call system when the call is answered at the wireless phone, regardless of the call priority.
Baxter Healthcare Corporation on 5/21/2022. Voluntary: Firm initiated
VOLUME OF PRODUCT IN COMMERCE
Nationwide and Canada