Company: SIGHT DIAGNOSTICS LTD
Date of Enforcement Report: 4/27/2022
The Sight OLO device is a computer vision based platform for blood analysis. The platform combines computer-vision algorithms for image processing to identify and quantify blood components (e.g., red blood cells) and their characteristics (e.g., cell volume) in an automated fashion. Using dedicated staining, the proposed platform provides complete blood count analysis. The Sight OLO is a compact device, designed to be automated and simple to operate, to enable rapid testing and analysis. The Sight OLO consists of a scanning and analyzing device and a CBC test kit, including disposable cartridges and sample preparation tools. The disposable cartridge containing the blood sample is loaded into the device through the loading slot. The device is operated through the touch screen interface. The Sight OLO provides complete blood count information with 5-part differentials for white blood cell types. Specifically, the CBC parameters measured by the Sight OLO are listed below and include: WBC, RBC, HGB, HCT, MCV, MCH, MCHC, RDW, PLT, NEUT%/#, LYMPH %/#, MONO %/#, EO%/# and BASO%/#. In addition, the Sight OLO signals specific WBC abnormal cases by flagging the sample.
In instances where custom reference ranges were configured on the device post installation, a possibility to inadvertently apply changes to reference range values was found on software versions 2.59.3 and all earlier versions, which can potentially lead to displaying and printing incorrectly-configured reference range values.
SIGHT DIAGNOSTICS LTD on 4/27/2021. Voluntary: Firm initiated
VOLUME OF PRODUCT IN COMMERCE
322 units: U.S 109 Foreign 213
U.S. Nationwide and International