Company: Siemens Healthcare Diagnostics, Inc.
Date of Enforcement Report: 8/12/2020
DCA Vantage Handheld Barcode Scanner – Zebra Model – Model # DS4308 – HC0015BZZWW external accessory to DCA Vantage Analyzer US
Recall Number: Z-2747-2020
If DCA Vantage Analyzer is configured to run Code 39 with check digit, the scanner is not confirming the integrity of read barcode data using the check digit. Also, if DCA Vantage Analyzer is configured to run Code 39 with check digit and an operator erroneously scans a Code 39 barcode without a check digit, the scanner incorrectly sends the code and the last character is removed.
Siemens Healthcare Diagnostics, Inc. on 6/30/2020. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE