Company: Braemar Manufacturing, LLC
Date of Enforcement Report: 3/26/2020
Braemar Manufacturing LLC DL900 Holter Monitor 1285 Corporate Center Dr. Suite 150 Eagan, MN 55121 USA Rx Only.
- Recall Number: Z-1459-2020
An issue with the DL900 and DL950 Series Holter Recorder was recognized on January 1, 2020. This issue can prevent the recorder from operating normally. Braemar confirmed that beginning on January 1, 2020 if a AAA battery is inserted in the recorder and a user attempts to start it, the recorder will display Error: 602 and fail to function for a new patient study. There are no actions that a clinical user can take to clear this error. Even if this error code is displayed, existing ECG recordings stored on the device can still be downloaded via the standard Holter Application Software. Error 602 may be able to be cleared by technical or engineering staff at the customer site.
Braemar Manufacturing, LLC on 1/22/2020. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
US Nationwide and International