Company: Zimmer Biomet
Date of Enforcement Report: 11/7/2019
The ROSA Brain device is a robotic platform that assists neurosurgeons in positioning medical instruments or implants during surgery. The device is composed of a compact robotic arm and a touch screen mounted on a stand. Different types of instruments may be attached to the end of the robotic arm depending on the procedure to be completed.
- Recall Number: Z-0172-2020
Zimmer Biomet recalled the ROSA Brain Device due to a software issue with ROSA Brain v184.108.40.206 (v220.127.116.11 software) and ROSA Brain v18.104.22.168 (v22.214.171.124 software, collectively referred to as v3.0 software), which can drive the robotic arm to an incorrect position resulting in risks for the patient.
Zimmer Biomet has received five complaints related to this issue, including one patient injury. No deaths related to this issue have been reported.
MEDTECH SAS on 9/10/2019. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide and International