Company: Philips Medical Systems (Cleveland) Inc
Date of Enforcement Report: 7/31/2019
BrightView SPECT, Model Number 882480
Recall Number: Z-2009-2019
An issue with the handcontroller may result in either spontaneous uncommanded motions or continued motion after the buttons were released.
Philips Medical Systems (Cleveland) Inc. on 6/10/2019. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
US Nationwide and International