Company:RAYSEARCH LABORATORIES AB
Date of Enforcement Report 7/25/2018
RayStation Radiation Therapy Treatment Planning System, Model nos. 2.5, 3.5, 4.0, 4.3, 4.5, 4.7, 4.9, 5.0, 6.0, 6.1, 6.2, 7.0 Product Usage RayStation is a software system designed for treatment planning and analysis of radiation therapy.
Recall Number: Z-2554-2018
The firm has learned that some RayStation/RayPlan users have commissioned machines with erroneous Beam profile correction parameters. These parameters affect the dose calculated in corners of large or off-axis fields. This effect cannot be seen in the Beam Commissioning module and dose in large or off-axis fields needs to be validated using the Beam 3D Modeling module in RayPhysics/RayPlan Physics. The user must be aware to avoid incorrect dose calculations during treatment planning.
RAYSEARCH LABORATORIES AB, Stockholm, Sweden on 6/6/2018. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE