Company:Tosoh Bioscience Inc
Date of Enforcement Report 8/22/2018
NG8 Automated HPLC Analyzer: HLC-723G8-ST, 021560; and HLC-723G8-LA, 021674 Product Usage: The Tosoh Automated Glycohemoglobin Analyzer HLC-723G8 is intended for in vitro diagnostic use for the quantitative measurement of % hemoglobin A1c (HbA1c) (DCCT/NGSP) and mmol/mol hemoglobin A1c (IFCC) in whole blood specimens. This test is to be used as an aid in diagnosis of diabetes and as an aid in identifying patients who may be at risk for developing diabetes.
Recall Number: Z-2745-2018
HbAE is known to interfere with the HbA1c assay on the current version of software, Ver. 5.23. Customers should exercise caution when reviewing chromatograms and ensure that Flag 43 is enabled on their device to avoid reporting invalid test results in the presence of HbAE..
Tosoh Bioscience Inc, Grove City, OH on 6/6/2018. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide and Internationally