Company:Ion Beam Applications S.A.
Date of Enforcement Report 7/18/2018
Proteus 235, Version PTS-18.104.22.168 and PTS-11.0.2 Product Usage: A device designed to produce and deliver a proton beam for the treatment of patients with localized tumors and other conditions susceptible to treatment by radiation.
Recall Number: Z-2424-2018
Correction vector confirmation message is lost if access point is changed after sending the correction vector. As a result, the patient will be treated in the setup position or treatment position instead of the corrected position.
Ion Beam Applications S.A, Louvain La Neuve . on 5/16/2018. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
US Nationwide in the states of FL and the countries of Sweden, The Netherlands