Company:ICU Medical Inc
Date of Enforcement Report 11/22/2017
Plum 360 Infusion System, List number 30010.
Recall Number Z-0101-2018
(1) Under certain conditions, if a malfunction alarm occurs while the pump is in the “Paused” state waiting for the distal pressure to decrease, the pump cannot be turned off and delivery cannot be restarted until the battery is discharged or is disconnected; and (2) when the user accesses the Preventive Maintenance Screen in Service/Biomed Mode with a Total Delivery Time >1,500 hours, the user will not be able to interact with the device and the device must be power cycled.
ICU Medical Inc., Lake Forest, IL on 10/30/2017. Voluntary firm initiated recall is ongoing
VOLUME OF PRODUCT IN COMMERCE
Distribution was nationwide to medical facilities. Foreign distribution was made to Canada and Australia. Government and military distribution was also made.