Date of Enforcement Report 4/192017
VITROS Immunodiagnostic Products NT-pro BNP Reagent Pack, REF/Catalog 680 2156, IVD — Ortho-Clinical Diagnostics Pencoed, Bridgend CF35 5PZ, UK
Recall Number Z-1765-2017
Increased frequency of calibration failures for VITROS Immunodiagnostic Products NT-proBNP Reagent Lots 1568, 1570, 1580 and 1590, due to background signals for these affected NT-proBNP lots that have been increasing with time since release testing, resulting in higher than expected VITROS Immunodiagnostic Products NT-proBNP level 1 calibrator signals produced, leading to calibration failures.
ORTHO-CLINICAL DIAGNOSTICS, FELINDRE MEADOWS ,Bridgend, United Kingdom on 3/7/2017. Voluntary: Firm Initiated recall is ongling.
VOLUME OF PRODUCT IN COMMERCE
US: 6324; Foreign: 1868
Nationwide and Internationally