Company: Philips Medical Systems
Date of Enforcement Report 2/10/2016
DS/US Proton Feature with Pinnacle3 Software Version 10.0.0 and 14.0.0 Radiation Therapy Planning System Model #459800200621
Recall Number Z-0745-2016
In DS/US proton planning, there is a correction factor used internally to the dose engine that is calculating improperly in some cases. This can result in an incorrect dose calculation.
Philips Medical Systems, Andover, MA on 1/8/2016. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
US: Nationwide Distribution in the states of FL, OH, and MO.