Company: GE Healthcare.
Date of Enforcement Report 11/18/2015
GE Healthcare, SIGNA Excite 3.0T. MR System for use as a diagnostic imaging device.
Recall Number Z-0256-2016
When performing head or neck scans, the currently displayed SAR values could be lower than the actual SAR in the head as predicted by SAR modeling. The predicted HEAD SAR value from the modeling suggests the actual SAR in the head could exceed IEC60601-2-33 limit of 3.2 W/kg for some scans.
GE Healthcare, Waukesha, WI. on 8/31/2015. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide and Internationally