Company: Mindray DS USA, Inc. dba Mindray North America
Date of Enforcement Report: 6/3/2015
Panorama Patient Monitoring Network, Multi- Parameter Patient Monitor (with Arrhythmia Detection and Alarm. Part number 0998-00-0708-01. The Panorama Network includes the Panorama Telemetry System, which acquires and monitors physiological data for ambulating patients within a defined coverage area.
Recall Number Z-1675-2015
Software anomaly in the Panorama System software version 8.9 that manifests when the Panorama is in use with Mindray Passport V Monitor. Arrhythmia alarms which were previously displayed on both the Passport V and Panorama will not be displayed on the Panorama following a restart of either system (due to communication loss, restart, etc.).
Mindray DS USA, Inc. dba Mindray North America, Mahwah, NJ on 3/12/2015 Voluntary: Firm Initiated recall is ongoing
VOLUME OF PRODUCT IN COMMERCE