Company: Philips Medical Systems, Inc.
Date of Enforcement Report 5/27/2015
Philips Healthcare DigitalDiagnost System X-Ray
Recall Number Z-1554-2015
The system is designed to emit a beep upon termination of an exposure. However, if the system has been powered on for over 12 hours, the system will no longer emit this signal. This is a failure to comply with 21CFR 1020.31(j).
Philips Medical Systems, Inc., Andover, MA 2/9/2015. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE