Date of Enforcement Report 4/24/2013
Alaris PC unit model 8015 with software version 9.12 Product Usage: The Alaris PC unit is the central programming, monitoring and power supply component for the Alaris System. Recall Number Z-1112-2013
The recall was initiated because Carefusion has identified potential risk associated with bolus programming with the Alaris PC unit (model 8015) software version 9.12. Automating infusion pump parameter input from either the electronic medical record (EMR) or the Alaris Auto-ID module for a continuous infusion with a bolus dose option may result in a loss of information from the Guardrails Data set while programming a bolus dose.
Carefusion Corporation, San diego, CA on 3/6/2013. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
4,090 total units
Nationwide and Internationally