Company:Varian Medical Systems, Inc.
Date of Enforcement Report 5/30/12
TrueBeam and True Beam STx versions 1.0 through 1.5, Model number H19; TrueBeam Radiotherapy Delivery System is intended to provide stereotactic radiosurgery and precision radiotherapy for lesions, tumors and conditions anywhere in the body where radiation treatment is indicated. Recall # Z-1175-2012
An anomaly has been identified with the respiratory gating software of TrueBeam. When importing breath-hold gating protocols, the gating thresholds can be reset to default values, rather than retaining the thresholds established during planning.
Varian Medical Systems, Inc. Oncology Systems, Palo Alto, CA, by letter on January 30, 2012. Firm initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide and Internationally