Company: Impac Medical Systems Inc
Date of Enforcement Report 12/1/2010
Elekta Impac Software Sequencer Verify & Record System, for MOSAIQ, The intended use of Sequencer (generically referred to as a verification system) is to assist in the process of process of patient treatment on a radiotherapy treatment machine. Recall # Z-0159-2011
Mistreatment– The database conversion utility used during the upgrade errantly changes the start gantry angle to zero. Any affected treatment fields, if not already treated, would start delivering from zero instead of the original prescribed value.
Impac Medical Systems Inc., Sunnyvale, CA, by letter dated February 19, 2010. Firm initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide, Australia, Belgium, France, Germany, Hong Kong, Ireland, Israel, and Italy