Company: Datascope Corp.
Date of Enforcement Report: 9/28/05
Panorama Patient Monitoring Network, Cardiac Arrhythmia Monitor. Part Number 0040-00-0366-20, Recall # Z-1567-05
The software version contains an anomaly which could freeze the unit when the right click button is used on the mouse.
Datascope Corp, Mahwah, NJ, by letter on August 19, 2005. Firm initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
Nationwide and Internationally