4/16/99 Class II Colleague 3 Volumetric Infusion Pump (Triple Channel). Baxter Healthcare 6,992 units. A software communication error occurs predominately when all three channels are in use, causing an alarm codition that will stop the function of all of the channels in use. The pump will alarm appropriately, alerting the user of the failure state. Z-872-9.