CIBA Corning Acs:180 and ACS:180 Plus Analyzers, used to conduct various laboratory diagnostictests. Recall #Z-368-9.
Due to a software error, the ACS Folate and ACS T-Uptake assays may run without the required conditioning solution. As a result, the test results may be outside of the system requirements.
All serial numbers for the ACS 180 and ACS 180 Plus Analyzers.
Chiron Diagnostics Corp., Oberlin, Ohio.
Manufacturer, by letter dated June 14, 1996. Firm-initiated recall complete.
Nationwide and international.
1,015 units were distributed.